to reading the User’s Manual. Please read the Accutrend. Plus User’s Manual carefully before using the meter for the first time. The Accutrend Plus meter is used. The Accutrend Plus system, one meter delivering both total cholesterol and glucose test results, can with Accutrend Plus may help employers improve employee productivity, improve .. 15 Cholestech LDX System User Manual. Inverness. View and Download Cobas Accutrend Plus user manual online. Accutrend Plus Blood Glucose Meter pdf manual download.
Roche Accutrend Plus Meter is a flexible, hand-held point of care device for the main risk factors of cardiovascular diseases (CVD). This device provides convenient and rapid detection of cholesterol and glucose. It is tailored to be used outside the lab in the physician’s office and in the hospital as well as for self-measurement at home or during sports. The Accutrend Plus instrument The Accutrend Plus instrument is used for quantitative measurement of 2 blood parameters: glucose and cholesterol. The instrument is suitable for professional use as well as for self-testing. If you have any questions which are not answered by this manual, call the ACCU-CHEK Customer Care Service. Accutrend® GC is also suitable for self-testing. With Accutrend® Glucose you can determine your blood sugar in only 12 seconds and with Accutrend® Cholesterol you can determine your cholesterol level in 180 seconds. The instructions below take you step by step through the meter’s operation, cleaning and care. Please take time to read.
|Published (Last):||12 October 2014|
|PDF File Size:||16.77 Mb|
|ePub File Size:||11.20 Mb|
|Price:||Free* [*Free Regsitration Required]|
Aug 29, 2018 Product Details Accutrend Plus Meter for lactate, cholesterol, glucose aaccutrend triglycerides Ready-for use anywhere: Measures 4 parameters Cholesterol, Triglyceride, Glucose, Lactate Stores up to measurements per parameter. The instrument temporarily shows all accutrend plus manual that may appear accutrend plus manual the display.
Starting The Measurement Performing a measurement Starting the measurement 13 Close the measurement chamber flap. Starting The Measurement Quality control recommendation Starting the measurement 11 Close the measurement chamber flap. Page 10 Contents Product specifications Operating plis and technical data Page 93 Memory After the usual display test has been performed when switching on the instrument, the last stored measured value is displayed.
COBAS ACCUTREND PLUS USER MANUAL Pdf Download.
Always power off the instrument before cleaning it! Page 73 Performing a measurement The time needed to evaluate the sample varies depending on the test parameter.
Preparing To Perform A Measurement Performing a measurement Preparing to perform a measurement Get the test strip container for the mmanual measurement. Performing a measurement Performing a measurement After it is switched on and coded, the instrument expects you to insert the test strip. Open the measurement chamber flap. Make sure the code strip belonging to these test strips is at hand unless the instrument has already been coded for this test strip lot.
Quality control recommendation Display of results When measurement is finished, the result is displayed. Solution Remove and discard the strip.
Performing a measurement Brief overview of the steps to be carried out Code the instrument once Insert the test strip Apply blood per test strip lot. The batteries are potentially hazardous also and should be disposed of accordingly. Solution Close the measurement chamber flap. Press the M button to change the currently flashing setting. Flap open displayed during measurement The measurement chamber flap is not closed properly.
If you plu the cap off the container of test strips.
Where illustrations are used, they always appear on the left side with the accompanying explanation on the right. The application area must be completely covered mmanual blood example Aotherwise incorrect values may be obtained.
The instrument is suitable for professional use as well as for self-testing. Note that all handling of blood samples represents a risk of infection. Memory Press the Set button to switch between the memory areas of the two test parameters.
Do not lance yourself before being instructed acccutrend do so in the course of this description. Page 96 Memory Special cases: Make sure that your assignments are unique and reproducible. If the beeper is enabled, you will hear a beep in the following situations: Press the M button until the desired day is displayed.
Regularly check that all display elements are functioning correctly to prevent misinterpretations due to a defective display. Page 79 Performing a measurement Assigning pluss information to a measurement: Test Principle Introduction Test principle By means of a code strip, the instrument reads the lot-specific characteristics of the test strips currently in use.
Accutrend Plus Cholesterol Meter User Manual 2017
Cobas Accutrend Plus User Manual
Performing a measurement Checking the applied blood: Page 9 Contents Quality control recommendation Preparing to perform a control test Medical staff and other people using the Accutrend Plus accutrfnd to perform measurements for more than one patient must be aware that any object coming into contact with human blood is a accitrend source of infection. Check that the display functions correctly by comparing it to the diagram on the left to prevent misinterpretations due to defective display elements.
Cleaning Cleaning A clean optical measuring system is a basic prerequisite for determining precise measured values.